In this post, I will look at the impact modifying mobile phase pH can have on reversed-phase separations.
Continue reading Using pH to optimize reversed-phase flash chromatography separations
Have you ever run flash column chromatography with mass detection (Flash-MS) and observed the total ion current or TIC increase during the purification only to find that there was no discernible compound contributing to the effect?
In this post I discuss how I came across this issue and the solution I found to work.
Continue reading How do I remove an annoying MS TIC background?
You have performed your synthesis and now it is time to purify the reaction mix. You have used thin-layer chromatography (TLC) and see a separation but when you try to purify with flash column chromatography, you can’t get the target compound separated from an impurity. So, what is happening (or isn’t happening)?
In this post I will give some input on why some separations are not transferrable from TLC.
Continue reading Why can’t I reproduce my TLC separation using flash column chromatography?
Myclobutanil is a fungicide of considerable concern to the cannabis industry. Its removal from extracted cannabinoid oil is the hot topic among processors and extractors. In today’s post we will present how flash chromatography can be used for the remediation of the pesticide, thereby significantly reducing it from the final product in a single step.
Continue reading Pesticide remediation of Cannabis oil- Myclobutanil removal by flash chromatography
An interesting question, at least to me. Depending on the detector brand, some mass spectrometer manufacturers recommend acetonitrile while others recommend methanol. Is there a real difference between these solvents?
In this post I look at how acetonitrile and methanol compare when used with an APCI source.
Continue reading Does mass detection make-up solvent choice matter?
I am always grateful for the feedback I get from my blog readers. Today’s blog is in response for multiple requests for tips on purifying complex mixtures and suggestions for alternative sample loading techniques.
In this post, I will attempt to address both, to some degree anyway, with a single example using a scavenger resin.
Continue reading In-line scavenging simplifies flash column chromatography reaction mixture purification
For chemists preferring or needing to dry load their crude sample mixtures to get an acceptable flash purification result, using the right ratio of sample to sorbent can be quite important. Too much sample and solubility issues can ensue, too little sample and significant band broadening occurs, reducing the separation quality.
In this post, I propose an acceptable ratio range based on my own experimental data.
Continue reading What is the optimal sample to sorbent ratio for dry loading in flash column chromatography?