To keep the evaporation and solventtrap in good working condition, the following items should be checked and replaced per the recommended frequency.
Tags: Manuals & User Guides
There are many things that can cause your Speed Vap III to have a problem with the vacuum. Such as the following examples:
1. The compressed air has to flow through the Speed Vap, if there is no exhaust, there is no vacuum. (The exhaust contains solvent vapors and should be properly vented)
2. There may not be sufficient compressed air being delivered to the Speed Vap. (2 CFM is required)
3. If the unit is connected to a Solvent Trap, the rubber stopper that goes into the collection flask may not be making a good seal.
4. The seal on Speed Vap III lid compresses over time and it will need to be replaced to maintain a good seal.
Tags: Tech Tips
Our Service Engineers have collected their best pro tips on this page for looking after your SPE-DEX, DryVap or SpeedVap equipment.
The Speed-Vap® Solvent Evaporation System provides safe and fast evaporation of n-hexane solvent for Oil and Grease or fat extractions.